pH Measurement Panca Mudji Rahardjo, ST.MT. Electrical Engineering -UB 1. Definition of pH In its most common interpretation, pH is used to specify the degree of acidity or basicity of an aqueous solution. Historically, pH was first defined as the negative logarithm of the hydrogen ion concentration (pondus Hydrogenii, literally hydrogen exponent), to simplify the handling of the very small concentrations
(on the order of 107 moles liter1 ) encountered most commonly in nature. This definition of pH is expressed as Equation below, where [H+] is the molar concentration of solvated protons in units of moles per liter. This led to the adoption of the more rigorous definition of pH as the negative logarithm of the hydrogen ion activity in solution, where a H+ is the hydrogen ion activity and is the activity coefficient
1. Definition of pH 1. Definition of pH In practice, the routine measurement of pH is not accomplished by the direct determination of the hydrogen ion activity. Rather, pH is determined relative to one or more standard solutions of known pH. Water, in the absence of other chemicals that might alter its equilibrium, contains 107 molar hydronium ions [H+] and 107 molar hydroxonium ions [OH] at 25 C.
Under these idealized conditions, water has a pH of 7.0, and is said to be neutral. 2. Electrochemical Methods of pH Measurement Electrochemical measurement of pH utilizes devices that transduce the chemical activity of the hydrogen ion into an electronic signal, such as an electrical potential difference or a change in electrical conductance. The Glass Membrane Indicator Electrode
The most widely used method for measuring pH is the glass membrane electrode. The Glass Membrane Indicator Electrode The Glass Membrane Indicator Electrode The construction of a typical pH indicator electrode is illustrated in Figure
71.1(b). The glass membrane at the tip of the electrode, which acts as the transducer of pH, is approximately 0.1 mm thick. One side of the membrane contacts the sample while the other contacts the electrode filling solution, an electrolyte of defined composition and pH. A reference element (e.g., a silver chloride coated silver wire) immersed in the filling solution makes a stable electrical contact between the potential measuring circuitry of the pH meter and the electrolyte in contact with the inner side of the glass membrane.
The reference electrode illustrated in Figure 71.1(c) provides a means of making electrical contact between the meter and the sample that is in contact with the external side of the indicator electrodes pH-sensitive glass membrane. Like the indicator electrode, the body of the reference electrode is filled with an electrolyte into which a reference element is immersed. The electrode also has a means, such as a porous ceramic frit, by which the reference electrode filling solution can make physical and electrical contact to the sample.
The importance of this liquid junction cannot be understated, as it is a potential source of sample contamination, measurement errors, and reliability problems. Glass Membrane Indicator Electrode Construction The hydrogen-ion selective glasses used to construct pH indicator electrodes are formed by fusing silica, alkali metal oxides, and alkali earth oxides. The silica component, SiO2, makes up approximately 70% of the glass.
The alkali metal oxide contributes mobile ions such as sodium or lithium, which act as electrical charge carriers, or are exchanged for protons in the hydrated glass layer. Components such as calcium oxide (CaO) are added to incorporate multivalent ions into the glass; these act to modify the network structure of the glass, imparting characteristics such as processability and chemical resistance. Glass Membrane Indicator Electrode Response to
pH When the membrane glass is immersed in aqueous solution, a gel-like hydrated layer on the order of 100 nm thick forms at the surface of the glass. Within this hydrated layer, the exchange of sodium ions in the glass for hydrogen ions in the solution produces a potential difference between the hydrated layer and the solution which depends on the hydrogen ion activity.
The net potential developed across the entire glass membrane, net, is the sum of boundary potentials, b, at the inner (reference element side) and outer (sample side) extents of the membrane. Reference Electrodes A reference electrode, such as that pictured in Figure 71.1(c), is used to make a stable, low resistance electrical contact between the external measuring circuit and the sample, establishing a reference potential against which the indicator electrode can be referred.
In the metallic conductors of the pH meter circuitry, current is carried by electrons, while in the sample electrolyte current is carried by ions. The difference between reference electrodes is primarily the oxidation or reduction reactions that effect charge transfer across the reference element/electrolyte interface. The composition of the reference electrode filling solution is, in general, different from that of the sample, and the physical contact between the two dissimilar electrolytes forms what is known as a liquid junction. The two phase boundaries (reference element/filling solution and filling
solution/sample) necessary to make electrical contact to the sample each introduce an additional potential, which adds to that of the indicator electrode, but proper design of the reference electrode ensures that these potentials remain constant and can be calibrated out. Reference Elements The most widely used reference electrode for pH measurement is the silver/silver chloride electrode. The electrode is constructed from a silver wire that has been coated
with silver chloride and immersed in an electrolyte saturated with silver chloride. Current is readily passed across the electrode/electrolyte interface by the reduction of silver chloride to form silver metal and chloride ion. Another commonly used reference electrode is the calomel electrode, based on the reduction of mercuric chloride to mercury. Liquid Junctions
The necessary contact between the filling electrolyte of the reference electrode and the sample forms a liquid junction, and results in the development of a junction potential that depends on the compositions of the two electrolytes. The stability of the liquid junction potential depends on the constancy of the interface between the filling electrolyte and the sample. Selection, Use, and Care of Reference Electrodes
Reference electrode selection is dictated by the sample to be studied and the measurement conditions, such as temperature. Silver/silver chloride electrodes are preferred over calomel for generalpurpose use, and at high temperatures (70C or greater). Calomel electrodes are preferred for high-precision determinations, and where silver contamination of the sample presents a problem. The type of liquid junction and its flow rate is important; a high flow rate may contaminate the sample and deplete the electrode, while a low flow rate may lead to clogging. Viscous, semisolid samples or low ionic strength samples require a high
flow rate junction. The reference electrolyte fill port should remain open during use of the electrode to insure adequate flow through the junction. Leaving the fill port open at all times will prevent any contaminants from entering the body of the reference electrode and reduce the likelihood of junction clogging, at the expense of more frequent refilling. Low maintenance reference electrodes use a polymer gel saturated
with electrolyte within the body of the electrode, and will perform properly if stored in filling solution. Calibration The glass indicator electrode is calibrated using standard buffer solutions of known pH. A two point calibration procedure is illustrated schematically in Figure 71.3. A pair of standards is chosen to bracket the pH range of interest.
Temperature Compensation The potential developed across the pH indicator glass electrode membrane is temperature dependent, with a temperature coefficient of approximately 0.3% perC, as follows from Equation 71.6. The effect of temperature on the electrode calibration is illustrated in Figure 71.3. Most pH meters have provision for temperature compensation, which corrects the slope of the measured potential versus pH calibration. Manual compensation permits the user to dial in the temperature at
which the measurement is going to be made. Meters equipped with automatic temperature compensation (ATC) use a platinum resistance thermometer to directly measure the temperature of the sample. The instrumentation within the meter then corrects the calibration function such that the millivolt reading is correctly interpreted as the pH of the sample at the measurement temperature. Great care is necessary in the use of the glass electrode type of pH probe.
Firstly, the indicator electrode has a very high resistance (typically 108) and a very low output. Hence, the output signal from the probes must be electrically screened to prevent any stray pick-up and electrical insulation of the assembly must be very high. The assembly must also be very efficiently sealed to prevent the ingress of moisture. A second problem with the glass electrode is the deterioration in accuracy that occurs as the glass membrane becomes coated with various substances it is exposed to in the measured solution.
Cleaning at prescribed intervals is therefore necessary and this must be carried out carefully, using the correct procedures, to avoid damaging the delicate glass membrane at the end of the probe. In some cases, careful brushing or wiping is adequate, whereas in other cases spraying with chemical solvents is necessary. Ultrasonic cleaning is often a useful technique, though it tends to be expensive.
Steam cleaning should not be attempted, as this damages the pHsensitive membrane. Mention must also be made about storage. The glass electrode must not be allowed to dry out during storage, as this would cause serious damage to the pH-sensitive layer. Finally, caution must be taken about the response time of the instrument. The glass electrode has a relatively large time constant of one to two minutes, and so it must be left to settle for a long time before the reading is taken. If this causes serious difficulties, special forms of lowresistivity glass electrode are now available that have smaller time
constants. pHFETs A relatively recent development in pH measurement is the introduction of systems based on the use of ion-selective field-effect transistors (ISFETs) as the sensing element. The integral amplification and small size have led to the development of inexpensive, battery-powered, pocket-sized pH measurement systems.
3. Optical Methods of pH Measurement For some applications, optical methods offer advantages over the use of the glass electrode or other electrochemical devices for the measurement of pH. Organic dye molecules with pH-dependent spectral properties have been routinely used for decades in acid-base titrations and in pH indicator papers. These dyes, many of natural origin, have more recently been put to use
as indicators to measure localized pH within living cells and in the development of fiber-optic probes for measuring pH within the body. Indicator Dyes Optical measurement of pH is based on the use of organic dye molecules that are weak acids or bases. The loss or gain of a proton changes the electronic structure of the molecule, producing a measurable change in the manner in which the molecule interacts with light.
This interaction can be the absorption of light at a particular wavelength, or fluorescence by one form of the molecule. Absorption Indicator Dyes The conjugate acid and base forms of absorption indicator dyes differ in the characteristic wavelengths at which they absorb light energy. A number of common absorption indicator dyes are listed in Table 71.3, and their pKas span the range of pH from approximately 1 to 13.
Fluorescent Indicator Dyes Fluorescent indicator dyes absorb light of a particular color (or wavelength) and re-emit some of the absorbed energy as light of a different color. The emitted light is at a longer wavelength than the exciting wavelength, and the difference is known as the Stokes shift. In principle, a fluorescence measurement is more sensitive than an absorption measurement because the only light measured by the detector originates from fluorescing molecules.
Table 71.4 lists a number of common fluorescent pH indicators used for measurement of pH in the physiological range. Fluorescein is widely used because the absorption maximum at 490 nm of the fluorescent dianion is readily excited by the 488-nm emission of argon ion lasers. Indicator Papers Indicator papers are a simple, rapid, and inexpensive means of measuring pH when the precision of an instrumental measurement is
not necessary. These are constructed from a strip of paper or plastic that has been impregnated with one or more absorption indicator dyes chosen to span the pH range of interest. The dyes are generally covalently attached to the strip, to prevent contamination of the sample by leaching of the dyes. A strip for indicating pH in the range of 5.5 to 9.0 can be constructed using the dyes bromocresol purple, bromothymol blue, and phenol red . The pH of a sample is determined to a precision of typically 0.5 units by
comparing the color of the strip to a color calibration chart provided by the manufacturer. Fiber-Optic pH Probes Optical pH sensors, often referred to as optrodes, represent some of the most sophisticated pH sensors, finding use for remote sensing in the body or industrial plants due to their small size and lack of electrical connections. They are typically constructed by immobilizing an indicator dye at the tip
of a light guide formed from one or more optical fibers, which are used to couple light between the indicator and the measurement instrumentation, as illustrated schematically in Figure 71.6(a). The resulting probes can be made very small, and with high-quality optical fibers, pH can be measured over considerable distances in electrically noisy environments that would interfere with potentiometrictype electrodes. Absorption optrodes measure the change in intensity of the light returned from the fiber tip/sensing region relative to the interrogating light of
specific (or a narrow band) wavelength. Light from a lamp was filtered to alternately select the l max of the base form of the dye (565 nm), and a wavelength (600 nm) where the absorbance is pH independent as a reference. The intensity of light scattered back into the second detector fiber was quantified by a photodiode. The ratio of the scattered intensities at the two wavelengths was then related to the pH of the sample.
Fluorescent indicator optrodes measure the Stokes-shifted fluorescence emission by the indicator and may use a single fiber to both interrogate and collect signal-carrying light. The amount of fluorescent pH indicator at the fiber tip must be maximized since fluorescence is emitted omnidirectionally and only a small fraction of the emitted fluorescence will be captured by the fiber. The source of excitation light is typically a high-intensity lamp or a laser. If a lamp, its output is filtered to select a narrow band of wavelengths at the absorption maximum of the specific form of the indicator, that is then introduced into the fiber to excite the fluorophore at the tip.
Emitted light collected by the fiber exits the fiber and is incident on a dichroic mirror, which reflects the long-wavelength light and passes shorter wavelengths. A filter then selects a narrow band of emission wavelengths, and finally, the light is detected. Due to the relatively small light intensities, the detector is typically a photomultiplier tube rather than a photodiode. References Measurement, Instrumentation, and Sensors
Handbook CRCnetBase, 1999. pp. 18681882 Alan S. Morris, Measurement and Instrumentation Principles. Butterworth. 2001
* The goal of our work is to design a router method that can provide isolation from unresponsive flows similar to what FQ does, as well as be simple to implement as RED. Of course we want to keep all...
Cystic Fibrosis and The Effects on the Pancreas By Jon Obert What is cystic fibrosis It is an genetic disease that is caused by a deletion of three base pairs on chromosome 7. This deletion causes a mutation in the...
The crew has to sail between Scylla (a six-headed monster) and Charybdis (a living whirlpool) As they are sailing, focused on Charybdis, Scylla's heads swoop down and eat six men "The Cattle of the Sun God"
IFRS 9 establishes principles for the financial reportingof financial assets and financial liabilities that will ... Cost Measurement. Objective of the business model ... Fair value hedge a hedge of the exposure to changes in the fair value of a...
Navigators looked through it at the sun or a star and determined its angle using the degrees marked on the instrument. Helped them determine the ship's latitude and local time. ... Goal to find a route to Asia. In 1488...
Much activity that takes place in a game has to be synchronized with the render cycle, so that one quantum of activity happens per frame. It doesn't work to move a character without regard to the frame rate.
Ready to download the document? Go ahead and hit continue!